Let’s begin by reviewing some terms that medical device reprocessing technicians should understand.
Let's start matching up another set of terms with their definitions. Ready?
Let’s hear from the
CPD Guy before
moving on to the
Let’s begin testing the cleaning efficacy of Washer #2 using the TOSI instrument washer test. Ready?
Place a TOSI and TEMP-170 on each test rack. Place one TEMP-90 on the middle
Place one wire test rack
into each instrument tray.
for Washer #2
Let’s hear from the CPD Guy while we wait for the cycle to complete.
Place one tray on each level of the washer rack for Washer #2:
The washer cycle is complete!
Let’s review the results of all three TOSI tests to see if they are a Pass or Fail.
Great job! Let’s investigate some problem areas of the washer to see what might be causing the failures.
Click on all of the problem areas of the washer until you find all nine!
Nozzle Not Aligned
Debris on Nozzles
Debris on Screen
Missing Spray Arms
Scaling On Wall
Dryer Vent Obstructing Spray Arms
Let’s hear from the CPD Guy once more before taking the quiz for your CEU!
Answer the ten true or false questions and fill in your contact information to earn your CEU